The portable MW100 Milwaukee PH Meter is a professional range measuring device but with an amateur price. It is very fast and precise and allows us to take comfortable readings directly in the nutrition tank of our crop. We can introduce the probe to go measuring the PH in real time as we incorporate acids and nutrients. With the addition of a probe, it is more difficult to damage the device due to water or humidity and the tasks are easier
The built-in probe has a cable of one meter in length so that we have sufficient maneuverability space and its plug is of the BNC type. It has a range of measurements between 0.0 and 14.0 with an accuracy of 0.1 pH.
The calibration is manual but very simple. It is done in two points, first with PH7.01 and then with PH 4.01. It is ergonomically designed to make it more comfortable and has a large LCD screen with very good resolution. It is a very fast and accurate meter thanks to its advanced and modern electronics.
The probe incorporates an electrode for taking measurements that is filled with an electrolite solution.
It is very suitable for use in indoor cultivation but has many more applications such as: aquariums, fish farming, education, laboratories, swimming pools, etc.
Instructions MW100 Milwaukee PH Meter
The first thing we have to do before using for the first time is to place the 9 volt battery included, taking care to respect the polarity
Before starting to take readings we must remove the protective cap of the electrode. If the last time we used it we would have left it dry we have to submerge it up to 2.5 cm in the washing liquid, also included for its reactivation.
We connect the probe to the meter
We make sure it is calibrated and if it is not calibrated for a long time, we do it before taking readings
Now, we introduce the probe into the liquid to measure, turn on the device and wait for the measurement to be stable.
When we have finished, we have to turn it off, take a few drops of the storage liquid to the electrode cap and cover it
MW Meter Calibration MW100
First calibrate with the envelope of pH7.01. We immerse the electrode in the liquid and wait for the reading to stabilize. With the black calibration button we must adjust it to pH7.0
Before calibrating to pH4.01 we must open the envelope and use a small amount to wash the electrode. Then we put it in the envelope and wait for the reading to stabilize. Finally we adjust with the PH4 / PH10 to PH4.0 button and with this we have calibrated it.
It is advisable to calibrate it at least once a month or after a long time if you use or when you change the electrode
Characteristics PH MW100
Range: 0.0 to 14.0
Resolution: 0.1 pH
Accuracy (at 25ºC): +/- 0.2 pH