The calibration procedure is simple and fast
Immerse the meter to the limit in EC calibration liquid (M10031)
Wait for the data to stabilize and with the screwdriver to the screw on the back,
Set to 1.4 (multiplied by 100 gives an EC of 1413). The meter is calibrated.
ALWAYS USE new calibration fluids.
Range? S 0-1990 / cm
Resolution 1 S / cm
Accuracy ▒ 2% of its capacity
Manual Calibration 1 point
Temp. Automatic compensation of 5 - 50 ░ C
Battery life 350 hours
Dimensions 150 x 30 x 24 mm